|Place of Origin:||China|
|Minimum Order Quantity:||N/A|
|Delivery Time:||2-4 Weeks|
|Payment Terms:||L/C, T/T|
|Supply Ability:||10 M tests/month|
|Name:||Hemoglobin (Hb) Test Meter With Accurate Results, Professional Use Only||Format:||Meter|
|Reading Time:||<15 Seconds||Pack:||2T/50T/1Set/100pcs/box|
|Storage Temperature:||-10 - 50℃||Shelf Life:||5 Years|
|Methodology::||Reflectance Photometer||Measurement Range::||4.5~25.6g/dL,45~256 G/L, 0.03-0.16 Mmol/L|
|Specimen Volume:||10 PL||Power Source:||3 AAA Batteries|
|Battery Life:||360 Hours Or 2700 Tests||Units Of Measure:||G/dU G/L, Mmol/L|
|Memory:||1000 Records||Automatic Shut Off:||6 Minutes After Last Action|
|Meter Dimension:||120mm*71mm*26mm||Display Size:||47*41mm|
Accurate Results Hemoglobin Test Meter,
Accurate Results Hb Test Meter,
Professional Use Hemoglobin Test Meter
Hemoglobin (Hb) Test Meter With Accurate Results, Professional Use Only
|Name:||Hemoglobin (Hb) Test Meter With Accurate Results, Professional Use Only|
|Reading Time:||<15 Seconds|
|Storage Temperature:||-10 - 50℃|
|Specimen Volume:||10 PL|
|Battery Life:||360 Hours Or 2700 Tests|
|Shelf Life:||5 Years|
|Measurement Range::||4.5~25.6g/dL,45~256 G/L, 0.03-0.16 Mmol/L|
|Power Source:||3 AAA Batteries|
|Units Of Measure:||G/dU G/L, Mmol/L|
|Automatic Shut Off:||6 Minutes After Last Action|
Hemoglobin (Hb) Test Meter with Accurate results, professional use only
1 Intended Use
The Hemoglobin (Hb) Test Meter is intended for the quantitative determination of hemoglobin and calculated hematocrit (HCT) in non-anticoagulated capillary whole blood or anticoagulated venous whole blood. The Hemoglobin Test Strip includes a mesh covered sample reaction zone. Specimen is applied to the center of the sample reaction zone. The hemoglobin is converted to methemoglobin to cause a color change on the strip. The meter reads the reflection of the strip every second until the end point of the reaction is detected. The reflection at the end point is directly proportional to the hemoglobin concentration.
2 Scope of Application
The Hemoglobin (Hb) Test Meter works with certain reagents strips manufactured by AllTest. for the quantitative detection. It’s for in vitro diagnostic professional use only. It may be used in emergency departments, clinical departments or medical services (such as community health centers), etc.
3 Product Name and Model Type
Product Name:Hemoglobin (Hb) Test Meter
4 Technical Specification
|Test Time||<15 seconds|
|Specimen Volume||10 μL|
|Power Source||3 AAA batteries|
|Battery Life||360 hours or 2700 tests|
|Units of Measure||g/dL, g/L, mmol/L|
|Automatic Shut Off||6 minutes after last action|
|Weight (without Batteries)||83g|
|Meter Connectors||USB cable for Data Transfer or Power (optional)|
5 Working and Storage Condition
|Parameters||Working Condition||Storage and Transport Condition|
|Altitude||Up to 2000m||Up to 2000m|
|Place||Indoor use, stable table, avoid of direct strong light and strong electromagnetic interference source||Clean, Ventilative, Stable|
Step 1: Unscrew the lancing device cover from the body of the lancing device.
Step 2: Insert a sterile lancet into the lancet holder and push it until stop in the lancet holder.
Step 3: Hold the lancet firmly in the lancet holder and twist the safety tab of the lancet, then pull the safety tab off. Save the safety tab for lancet disposal.
Step 4: Carefully screw the cover back to the lancing device. Avoid contact with the exposed needle. Make sure the cover is fully seated on the lancing device
Step 5: Adjust the puncture depth by rotating the lancing device cover. There are a total of 6 puncture depth settings. To reduce discomfort, use the lowest setting that still produces an adequate drop of blood.
Step 6: Pull the cocking barrel back to set the lancing device. A click may be heard. The device is now loaded and ready for obtaining a drop of fingertip blood.
Step 7: Prior to collecting the capillary blood specimen, make sure the patient’s hand is warm and relaxed. Massage the hand from the wrist up to the fingertip a few times to encourage blood flow. Clean the testing site with an alcohol swab and then dry the testing site thoroughly.
Step 8: Hold the lancing device against the side of the finger to be lanced with the cover resting on the finger. Push the release button to prick the fingertip. A click should be heard as the lancing device activates. Gently massage from the base of the finger to the tip of the finger to obtain the required blood volume
Step 9: Wipe away the first drop of blood. Apply light pressure to obtain a second drop of blood. Hold the tube slightly downward and touch the tip of the capillary transfer tube to the blood drop. Capillary action will automatically draw the sample to the fill line and stop.
Note: Make sure the blood covers the air vent of the tube or it will be hard to squeeze blood out. Never squeeze the capillary transfer tube while sampling.
Step 10: Unscrew the lancing device cover. Place the safety tab of the lancet on a hard surface and carefully insert the lancet needle into the safety tab. Press the release button to make sure that the lancet is in the extended position. Slide the ejection button forward to eject the used lancet. Place the lancing device cover back on the lancing device.
Note: Dispose of the used lancet according to local regulations.