|Kit Size:||25 T/Kit||Storage:||2-30℃|
One Step Rapid Test,
home testing kits
Convenient and High Quality Dengue IgG/IgM One step Easy Use Rapid Test Reader
The Dengue Rapid Test Cassette (Whole Blood/Serum/Plasma) is a rapid chromatographic immunoassay for the qualitative detection of IgG and IgM antibodies to Dengue virus in human whole blood, serum, or plasma as an aid in the diagnosis of primary and secondary Dengue infections.
Dengue is a flavivirus, transmitted by Aedes aegypti and Aedes albopictus mosquitoes. It is widely distributed throughout the tropical and subtropical areas of the world,1 and causes up to 100 million infections annually.2 Classic Dengue infection is characterized by a sudden onset of fever, intense headache, myalgia, arthralgia and rash. Primary Dengue infection causes IgM antibodies to increase to a detectable level in 3 to 5 days after the onset of fever. IgM antibodies generally persist for 30 to 90 days.3 Most Dengue patients in endemic regions have secondary infections,4 resulting in high levels of specific IgG antibodies prior to or simultaneous with IgM response.5 Therefore, the detection of specific anti-Dengue IgM and IgG antibodies can also help to distinguish between primary and secondary infections.
The Dengue Rapid Test Cassette (Whole Blood/Serum/Plasma) is a rapid test that utilizes a combination of Dengue antigen coated colored particles for the detection of IgG and IgM Dengue antibodies in human whole blood, serum, or plasma.
Allow the test cassette, specimen, buffer, and/or controls to reach room temperature (15-30°C) prior to testing.
1.Exact QR Code from the QR code card for specific information of the test. Use only QR code card provided inside the test kits.
2.Remove the test cassette from the sealed pouch and use it within 1 hour.
3.Place the test cassette on a clean and level surface.
>For Serum or Plasma Specimens:
To use a dropper: Hold the dropper vertically, draw the specimen up to the Fill Line (approximately 5 μl), and transfer the specimen to the specimen well of the test cassette, then add 1 drop of buffer (approximately 40ul) and start the timer. Avoid trapping air bubbles in the specimen well.
To use a micropipette: Pipette and dispense 5 µl of specimen to the specimen well of the test cassette, then add 1 drop of buffer (approximately 40 µl) and start the timer.
>For Whole Blood (Venipuncture/Fingerstick) Specimens:
To use a dropper: Hold the dropper vertically, draw the specimen about 1cm above the Fill Line, and transfer 1 drop of whole blood (approximately 10 µl) to the specimen well of the test cassette, then add 1 drop of buffer (approximately 40 ul) and start the timer.
To use a micropipette: Pipette and dispense 10 µl of whole blood to the specimen well of the test cassette, then add 1 drop of buffer (approximately 40 µl) and start the timer.
4.Wait for the colored line(s) to appear. The result should be read with AlltestR LF Reader at 10 minutes.
5.Do not interpret test results visually
6.For AlltestR LF Reader installation, start up and complete instructions refer to AlltestR LF Reader User Manual. Operator must consult AlltestR LF Reader User Manual prior to use and become familiar with the processes and quality control procedures.
Note: It is suggested not to use the buffer beyond 6 months after opening the vial.
INTERPRETATION OF RESULTS
Results read by Alltest TM LF reader
The result of positive or negative for Infectious Dengue IgG/IgM is determined by the Alltest TM LF reader.
Result could be exported to excel and PDF.